Jump to navigation
Jump to search
This article needs additional citations for verification . Please help improve this article by adding citations to reliable sources . Unsourced material may be challenged and removed. (December 2015) ( Learn how and when to remove this template message )
Basic diagram of a hollow-cathode lamp
Hollow-cathode lamps from an atomic absorption spectrometer
A hollow-cathode lamp (HCL) is type of cold cathode lamp used in physics and chemistry as a spectral line source (e.g. for atomic absorption spectrometers ) and as a frequency tuner for light sources such as lasers . An HCL takes advantage of the hollow cathode effect , which causes conduction at a lower voltage and with more current than a cold cathode lamp that does not have a hollow cathode. 
An HCL usually consists of a glass tube containing a cathode , an anode , and a buffer gas (usually a noble gas ). A large voltage across the anode and cathode will cause the buffer gas to ionize , creating a plasma . The buffer gas ions will then be accelerated into the cathode, sputtering off atoms from the cathode. Both the buffer gas and the sputtered cathode atoms will in turn be excited by collisions with other atoms/particles in the plasma. As these excited atoms decay to lower states, they will emit photons , which can then be detected and a spectrum can be determined. Either the spectrum from the buffer gas or the sputtered cathode material itself, or both, may be of interest. 
An HCL can also be used to tune light sources to a specific atomic transition by making use of the optogalvanic effect , which is a result of direct or indirect photoionization . By shining the light source into the HCL, one can excite or even eject electrons (directly photoionize) from the atoms inside the lamp, so long as the light source includes frequencies corresponding to the right atomic transitions. Indirect photoionization can then occur when electron collisions with the excited atom eject an atomic electron.
= atom in excited state, and
The newly created ions cause an increase in the current across the cathode/anode and a resulting change in the voltage, which can then be measured.
To tune the light source to a specific transition frequency, a tuning parameter (often the driving current ) of the light source is varied. By looking for a resonance on a data plot of the voltage signal versus source tuning parameter, the light source can be tuned to the desired frequency. This is often aided by use of a lock-in circuit.
See also[ edit ]
- List of light sources
- Electrodynamic tether#Hollow cathode
References[ edit ]
- ^ Eichhorn, H.; Schoenbach, K. H.; Tessnow, T. (1993). “Paschen’s law for a hollow cathode discharge” (PDF). Applied Physics Letters. 63 (18): 2481–2483. Bibcode : 1993ApPhL..63.2481E . doi : 10.1063/1.110455 . ISSN 0003-6951 . Retrieved June 5, 2017.
- ^ Analytical Methods for Atomic Absorption Spectrometry (Release E (2000) ed.). Perkin Elmer, Inc. 1964–2000. pp. 6–9.
- Atomic physics
- Optical devices
- Gas discharge lamps
- Articles needing additional references from December 2015
- All articles needing additional references
- This page was last edited on 29 April 2018, at 14:01 (UTC).
- Text is available under the Creative Commons Attribution-ShareAlike License ;
- About Wikipedia
- Contact Wikipedia
- Cookie statement
- Mobile view